For access to more articles, create an account using the "Sign Up" button
To view Biomomentum's Knowledge Base articles visible only to registered users, you must create an account. Click "Sign Up" in the top right of the screen to get started.
Stage does not move (Axis 1: Motor Not Enabled)
In the case where the stage does not move when "Up" or "Down" are clicked, it is likely that the stage has been turned OFF. To turn the stage back ON, click "Enable Axis" or "Enable Axes" in the "Manual Controls" window. Stages will automatically ...
DAQmx Self-Test (Error: -88705)
This error is usually caused by having an incorrect DAQ# in the Mach-1 Motion configuration file. Windows can sometimes change this number during a Windows Update. To fix this, the number can be changed in the Mach-1 Motion Configuration/Hardware ...
Load Cell Handling
It is important to be knowledgeable about your load cell before handling such a sensitive device. Please view the following video for basic tips on handling: https://youtu.be/ah3yxFmwBas Another particularity of single-axis load cells is that the ...
Basler Pylon Gain Out Of Range (Error code: 1172)
The solution to this error is to reduce the gain of the camera. SOLUTION: 1) Open Mach-1 Motion 2) Open camera (error occurs) 3) Open camera settings window 4) Reduce gain to <12. 5) Restart Mach-1 Motion [in one case it was necessary to end its ...